- Identifies Protein Residues on Surface
- Ideal for Hygiene Testing in Bakeries, Restaurants, Food Manufacturing Plants and More
- Interprets Results in 1-10 Minutes (No Additional Instruments Required)
- Allow 3 Business Days for Processing. NOTE: ONLY Ships on Monday, Tuesday and Wednesdays Due to Sensitive Contents
- FREE SHIPPING ($60 Value)
- As this product is temperature-sensitive, we cannot accept returns once this product leaves our facility.
- Only Ships Within the U.S. and Canada
ProClean instantly detects the presence of protein residue on a surface. You simply swab the surface you are testing and enact the test and wait 1 to 10 minutes for results. If the reagent does not change color, the surface does not have any protein residue. If the reagent turns purple, protein residue has been detected. The more contamination present, the quicker the reagent will turn purple and the darker that purple will be.
Hygiena's ProClean is ideal for:
- Food manufacturing plants
- Any place where potential contamination is a concern
Learn More About Our Complete Product Table
"I was delighted to speak with a person regarding the order and I will be a returning customer. Keep it up. Everything was much better than where I used to order refills."
"I have to admit that your services were Outstanding."
**Due to new restrictions, we are unable to ship this product outside of the U.S.
View Directions To Use Here
Directions for Use:
Allow the ProClean sample collection device to come to room temperature.
Remove the device from the foil pouch.
Write the location on the collection device in permanent marker.
Twist and pull the swab out of the tube of the collection device.
Swab a 4 inch x 4 inch area working horizontally (left to right) and then swab the same area again working vertically (up and down) so that you've swabbed it at least twice over. Make sure to swab with even coverage. If swabbing an uneven surface, keep the swabbing technique consistent and swab an area that is approximate with 4 x 4 inches.
The swab should bend while collecting the sample to ensure enough pressure is being applied during swabbing.
Do not touch the shaft of the swab while collecting, as this will contaminate the test. Hold it only by the end opposite the swab.
Make sure to rotate the swab while collecting.
Insert the swab back into the tube of the collection device and make sure it is sealed.
Holding the collection device in your fist, use your thumb to break the Snap Valve by bending it forward and then use your finger to bend it back to ensure it is completely broken.
Squeeze the bulb of the Snap Valve twice to ensure all the liquid inside goes into the tube.
Shake the entire tube for 5 to 10 seconds so the sample mixes with the liquid.
Hold the ProClean testing device upright while you wait for results.
The longer you wait, the more sensitive the ProClean test becomes.
ProClean can detect:
As little as 80 micrograms of contamination after one minute at room temperature
As little as 50 micrograms of contamination after five minutes at room temperature
As little as 20 micrograms of contamination after 10 minutes at room temperature
Positive - If the liquid inside the tube turns any shade of purple, protein residue has been detected.
Negative - If no color change occurs, no protein residue has been detected.